If an autoclave connected to the VistaCool system is displaying a cycle default notification, the exhaust port line is blocked.
- First, check for any kinks in the tubing that runs from the autoclave to the VistaCool manifold.
- Check to ensure that water is running from the tank coolant overflow line to the drain adapter.
- Make certain that the water supply valve is in the ON position. If there is no water available to cool the coil(s) in the tank, the inline thermal sensor located in the condensate line will respond by closing the pathway to alert you of the problem and to protect the plumbing from steam.